The Electro-Cap System makes QEEG preparation fast and easy. No individual electrode placements. No more plugging in individual electrodes. Electro-Cap System I comes with all of the basic accessories required for preparation of quantitative EEG. Lexicor / Neurofield Style EEG caps require ear lead attachments.
Electro-Caps are an EEG electrode application technique. They are made of an elastic spandex-type fabric with recessed, pure tin electrodes attached to the fabric. The electrodes are positioned to the International 10-20 method of electrode placement.
The medium Electro-Cap that is included with the Electro-cap System I will fit approximately 65% of all patients over the age of 5.
If you are not sure of the Electro-cap configuration that you need please contact us.
For use with Mindset, Discovery24, Lexicor,Deymed, Mitsar and other 19, 21 and 24 channel EEG systems.
Use only Ivory or Palmolive liquid detergent for washing your cap. Other soaps and detergents can leave a residue on the electrode surface, causing artifacts and/or impedance problems.
Use only ECI Electro-Gel. Other types of paste and gel can cause artifacts and/or high impedance.
Ensure that all accessory electrodes are the same metal as the electrodes in your cap. All standard ELectro-Caps have tin electrodes. Use of accessory electrodes of a different metal can cause a myriad of problems.
Keep all gel, paste, and moisture away from the connector that plugs into your equipment.