The ProComp Infiniti is an 8 channel biofeedback and neurofeedback system with the power needed for real-time data acquisition in any clinical or educational setting.
The ProComp Infiniti has eight protected pin sensor inputs with two channels sampled at 2048 sampes per second and six channels sampled at 256 samples per second.
The first two sensor channels of the ProComp Infiniti encoder provide ultimate signal fidelity for viewing raw EEG, EMG and EKG signals (2048 samples per second). The remaining six channels can be used with any combination of sensors, including EEG, EKG, RMS EMG, skin conductance, heart rate, blood volume pulse, respiration, goniometry, force, and voltage input.(256 samples/sec) Sensors are not included with this system.
ProComp Infiniti offers internal, user-activated calibration to ensure that you can always obtain the highest quality signal, without the costly downtime associated with factory re-calibration.
In short, the ProComp Infiniti covers the full range of objective physiological signals used in clinical observation and biofeedback.
Housed in an ergonomically designed case and requiring only a USB port, ProComp Infiniti can be used with any PC laptop or desktop. ProComp Infiniti can capture data in real time by connecting directly to the PC via fiber-optic cable, or it can store data on a Compact Flash memory card for uploading later to the PC.HARDWARE FEATURES
The BioGraph Infiniti software is created in a modular fashion to ensure you get only what you need and do not fill your computers memory with unwanted software. The Developer Tools will allow for high end users to create their own screens, scripts and channel sets. This ultimately provides the customer with the most variety under one system.
The ProComp Infiniti hardware comes with the BioGraph Infiniti software and the BioGraph Infiniti 360 Suite.
|FlexComp Infiniti||ProComp Infiniti||ProComp5||ProComp2|
|Compatible Sensors||EEG, SCP/EP, EMG, EKG, BVP(Blood Volume Pulse), Respiration, Temperature, Skin Conductance, Inclinometer, Algometer / Force, Bend, Foot Switch, Hand Push Button, Voltage Isolator, Force Adapter, Goniometer / Torsiometer Adapter, HEG||The same as FlexComp Infiniti||The same as FlexComp Infiniti||The same as FlexComp Infiniti|
|Sensor Type||External sensors||External sensors||External sensors||1 Internal EEG on Channel A. External sensors on B, C & D|
(with EEG-Z sensor)
|All channels||All channels||All channels||Channel B only|
|Sampling Rate||2048 samples/sec on all 10 channels||2048 samples/sec
on Channel A and B,
on Channel C, D, E, F, G, H
on Channel A and B,
on Channel C, D, E
on Channel A and B,
on Channel C and D
|Connection to PC||Fiber Optic to USB through TT-USB interface unit. Or Compact Flash Module Bluetooth Tele-Infiniti||The same as FlexComp Infiniti||The same as FlexComp Infiniti||Fiber Optic to USB (TT-USB) or COM port (Pro-SB) or Tele-Infiniti|
|Software and Suites||BioGraph Infiniti, Developer Tools, EEG Suite, Physiology Suite, and DynaMap Suite||BioGraph Infiniti, Developer Tools, either EEG Suite or Physiology Suite||The same as ProComp Infiniti||The same as ProComp Infiniti|
|Weight||200g (7oz)||200g (7oz)||200g (7oz)||40 g (1.4oz)|
|Size||5.1" x 3.7" x 1.5"
(130 mm x 95 mm x 37 mm)
|The same as FlexComp Infiniti||The same as FlexComp Infiniti||2.2" x 2.8" x 0.75"
(55 mm x 71 mm x19 mm)
|Power Source||4AA batteries, single-use alkaline or NiMH rechargeable||The same as FlexComp Infiniti||The same as FlexComp Infiniti||1 AA battery, single-use alkaline or NiMH rechargeable|
|Battery Life (alkaline)||30 hours typical, 20 hours minimum||The same as FlexComp Infiniti||The same as FlexComp Infiniti||10 hours typical|
View Computer Requirements for Thought Technology software (opens in a new browser tab)
1 Year manufacturer warranty.
General Requirements for Safety Compliant: UL60601-1 (R2003), CSA C22.2 601.1-M90 (R2005)
NOTE: ProComp Infiniti and its Sensor Heads and sensors are sensitive electronic instruments and should be handled as such. Be especially careful to avoid both pulling on the sensor cables and leads or getting moisture and gel on the sensor snaps. If necessary, wipe the surface with a damp cloth or use a moistened Q-tip to remove gel from inside the sensor snaps. Wipe with a dry cloth.